Purchase this article with an account.
Lapam Panda, Umasankar Barik, Suryasmita Nayak, Biswajit Barik, Gyanaranjan Behera, Ramesh Kekunnaya, Taraprasad Das; Performance of Photoscreener in Detection of Refractive Error in All Age Groups and Amblyopia Risk Factors in Children in a Tribal District of Odisha: The Tribal Odisha Eye Disease Study (TOES) # 3. Trans. Vis. Sci. Tech. 2018;7(3):12. doi: https://doi.org/10.1167/tvst.7.3.12.
Download citation file:
© ARVO (1962-2015); The Authors (2016-present)
To evaluate effectiveness of Welch Allyn Spot Vision Screener in detecting refractive error in all age groups and amblyopia risk factors in children in a tribal district of India.
All participants received dry retinoscopy and photorefraction; children also received cycloplegic retinoscopy. Statistical analysis included Bland-Altman and coefficient of determination (R2).
Photoscreener could not elicit a response in 113 adults and 5 children of 580 recruited participants. In Bland-Altman analysis mean difference of Spot screener spherical equivalent (SSSE) and dry retinoscopy spherical equivalent (DRSE) was 0.32 diopters (D) in adults and 0.18 D in children; this was an overestimation of hyperopia and underestimation of myopia. In Bland-Altman analysis of SSSE and cycloplegic retinoscopy spherical equivalent (CRSE) the mean difference was −0.30 D in children; this was an overestimation of myopia and underestimation of hyperopia. In regression analysis the relationship between SSSE and DRSE was poor in adults (R2 = 0.50) and good in children (R2 = 0.92). Cubic regression model for Spot versus cycloretinoscopy in children was: CRSE = 0.34 + 0.85 SSSE − 0.01 SSSE2 + 0.006 SSSE3. It was 87% accurate. Sensitivity and specificity of Spot in detecting amblyopia risk factors (2013 American Association for Pediatric Ophthalmology and Strabismus [AAPOS] criteria) was 93.3% and 96.9% respectively. Sensitivity of Spot screener in detection of amblyopia was 72%.
Photoscreener has 87% accuracy in refraction in children. Its value could be used for subjective correction tests.
Photoscreening could complement traditional retinoscopy to address refractive error in children in a resource-limited facility region.
This PDF is available to Subscribers Only